The frequency medicine software
(This item is presented for the first and will soon be available for sale)
This Windows compatible software (since the Vista version) can generate all the frequency combination (sequentially or simultaneously), or frequency modulation (with a chronology of variation in the case of isochronous frequencies). This signal must then be transmitted, via a good audio helmet, with the ears of the patient. This form of therapy is very powerful because it directly address to the brain.
The database includes already a very great number of elements which are contained in lists. Feel free to add to or modify.
- The simple and prefixed frequencies in database are based on the musical note (see correct value of the A musical note in documentation), the frequencies of Solfeggio, or on correspondences with chakras, planets, minerals, organs and systems.
- The modulated frequencies contained in the database are used for acupunctures points. With the MF1 or MF2 in biofeedback setting, it will be possible for you to detect the modulated frequencies which correspond to the person tested to make a treatment.
- The lists of frequencies are used mainly for the RIFE frequencies.
Possible types of modulation are:
- Amplitude modulation
- Frequency modulation
- Channel modulation
- Phase modulation
Types of signal transmission are:
- Simple sinusoidal frequencies,
- Frequencies modulated with its type of modulation,
- Lists of frequencies being able to be emitted into simultaneous or sequential,
- Audio signals (message subliminal, isochronous music of relieving, already recorded sounds,
- Progressive modulation to be able to create your own isochronous sounds,
- Noises: white, pink, brown.
A chronological mixing is realizable starting from the frequential editor present in the software.
A search engine allows you to search the database according to the list, the name and description that characterizes an item.
A utility saves a subliminal message and reverse so that only the unconscious part can decrypt.
The amplitude modulation (related with the isochronous sounds) is especially employed for power passing an infra low frequency to impose cerebral frequencies to obtain particular state of consciousness. The binaural sounds using a difference in frequency between the 2 ears are not implemented in the software because the effects are less compared to the isochronous sounds. In case you wish to impose a low frequency, the carrier frequency is less important. We can then set arbitrarily (between 400 and 1200Hz) or select a frequency from the musical notes in accordance with the principle of Nogier or harmonic (= octave) frequency modulation (multiple of 2). The value of the carrier frequency must be at least equal to 10 times the value of the modulation frequency. Amplitude modulation or frequency is monaural while channel modulation or phase modulation uses the stereo. Phase modulation is a modulation of the phase shift between two ears. It is located halfway between the modulation frequency and modulation channel.
In the concept of toning, use the amplitude modulation. In the concept of dispersion, use the frequency modulation (or phase modulation in some case). We will be able for example dispersing an acupuncture point and then another tone with the frequential editor. In the concept of harmonization, use the channel modulation. In some pathology of right/left synchronization, use channel or phase modulation.
To regulate a dissonance between two organs, two chakras or between two planetary aspects in astrology, you can for example send with the frequency editor in each earphone headset the base frequency of the element to solve this bad aspect.
Toning disperse or harmonize an element, use the base frequency resonance of the element and search the frequency modulation with a biofeedback (MF1 or MF2), then the type of modulation.
To find the resonant frequencies of a person, a biofeedback device (MF1 and MF2) is used. It is advisable to first seek the resonance frequency of the carrier (without modulation) and then search for the modulation frequency (we take the amplitude modulation type for the test). Then to the most suitable type of modulation. We then notice a very unstable reading of the measuring device, indicating that the parameters chosen are very reactive.
To download the demonstration version accessible by the heading “Support”, please register first on our Website.
To order it, go to the Shop section of the site.